Classification: Laboratory Thermostatic Devices
Model Number: J-HH-S.25
Heating power: 1000 W
Temperature display: Digital
Temperature range: RT~300 Celsius Degree
Precision: 1 Celsius Degree
Dimension: 350 * 310 * 330 mm
Workable area (mm): 250 * 250 mm, Round single hole
Capacity: 11 L
Material of chamber: 304 stainless steel
Single-hole Oil Bath Round Chamber 11L Φ 250 mm
Digital display constant temperature oil bath is widely used in drying, concentrating, distilling, impregnating chemical reagents, impregnating medicines and biological products. It can also be used in oil bath constant temperature heating and other temperature tests. An essential tool for environmental protection, medicine, hygiene, biochemical laboratories, analysis rooms, education and scientific research.
1. The working box is made of stainless steel, which has excellent corrosion resistance.
2. Precise temperature control, digital display, automatic temperature control.
3. Easy to operate and safe to use.
Main technical parameters
Temperature control form: digital display automatic temperature control
Temperature control range: room temperature–300℃
Temperature control accuracy: ±1℃
AL: Set the upper limit alarm value
HY: 0-50 , the original is 2
Proportional band P: 1–200℃, if it is 0, it is bit control, the original is 15
Integral I: 0-999 seconds
Differential D: 0-999 seconds
Period T: 1-200 seconds, the original is 90
Error value SC: sensor error correction value -50–50℃
Parameter lock LCK: 0 is not locked, 1 is to lock parameters other than the set value, 2 is to lock all parameter values
Heating power: 1000W
Studio size: diameter 250*250 mm, capacity is 11L
Dimension: 350 * 310 * 330 mm
Power supply: AC 110 – 240V, 50Hz/60Hz
This product is a tank type, tank made by stainless steel , cold-rolled steel chassis with high-quality well-made. Temperature section with a special electronic temperature control. The pot with digital display actual temperature, the product with beautiful shape and corrosion resistance, stability temperature, energy-saving and practical, easy maintenance, long life and other characteristics.
Using methods and precautions
1. Security of electricity: The product use of three holes safety outlet socket AC 220V power, please properly grounded, please do not ground to gas pipe.
2. Please inject water into the pot above bulkhead before plug the power. Please make sure do not out of the water when using heat pipe, otherwise it may cause burnning heating pipes, leak solder melt or electric shock, etc..
3. Setting power and temperature : Press the key of SET to check the point of temperature. Press button digital control of SET then the character starts to flash, indicating that the instrument into the setting, press the key to set the value increases , decreases. Press key to set the value, press key quickly again to change the instrument back to normal operation temperature when setting is completed. (If the temperature controller is intelligent, then attach the thermostat instructions ).
4. The machine has heating and thermostatic two light indicates working status , thermometers head showing the value of the actual temperature of the pot.
5. Please do not inject too much using water when 100 Celsius boiling distillation, to avoid overflow when boiling.