Quality stainless steel kettle body 304, using a circular tongue and groove seal, manually screw sturdiness. With good sealing, high safety factor, consume less acid solvent, easy to use!
1.Anti-corrosion, no harmful substances spill, reduce pollution, use of safety.
2.Heating, after boosting, can be dissolved rapidly destructive sample under conventional conditions is difficult and the sample containing the dissolved volatile elements.
3.Beautiful appearance, reasonable structure, simple operation, short analysis time, reliable data.
4.Inside a PTFE liner, generate care, acid and alkali.5.To replace the platinum crucible solve handling problems Dissolving trace element analysis of high purity alumina. High pressure digestion vessel used in the oven, also known as sample dissolution, pressure melting bombs, hydrothermal synthesis reactor, the digestion tank, PTFE high-pressure tank, is the determination of trace elements and trace elements sample digestion assistant. Sample pretreatment digestion of heavy metals, pesticide residues, food, mud, earth, fish, organic matter and so on.
Method of operation
The reaction was poured into a PTFE liner or liner within the PPL, and to ensure that the addition coefficient is less than 0.8.
Make sure the kettle under the pads in the correct position (convex side down), and then into the bush or PPL PTFE liner and the gaskets, tighten the kettle lid, then screw the kettle lid knob tight.
The reaktor sintesis hidrotermal placed in a heater, in accordance with a predetermined heating rate was raised to the desired reaction temperature. (Less than a predetermined safe temperature).
When confirmation of intra-abdominal temperature below the boiling point solvents reaction system can only open the kettle lid subsequent operations. Completion of the reaction will be when its cool, but also in strict accordance with the provisions of the cooling rate operation, to facilitate safety and reactor life.
After confirming the intra-abdominal temperature below the boiling point solvents reaction system, first screw the kettle lid release knob, and then turn on the kettle li
Hydrothermal synthesis reaction vessel promptly to each use clean, so as not to rust. Kettle body, kettle lid seal line to pay particular attention to clean and prevent damage to their bruised.