Classification: Laboratory Thermostatic Devices
Digital thermostat water bath pot is mainly used for the laboratory drying, distillation, concentration, and soaking the chemical or biological products, it can also be used for heating and other temperature test, it is a necessary tool of biological, genetic, viral, aquaculture, environmental protection, medicine, health, laboratory, analysis room, education and scientific research.
1. Water tank studio is made by stainless steel, with excellent corrosion resistance.
2. High precision temperature control, digital display, automatic temperature control.
3. Easy to operate and easy to use
4. HH-8 with magnetic stirring to accelerate water temperature evenly
Name: Digital display thermostat
Model: HH-1, HH-2, HH-4, HH-6, HH-8
Heating power: 300W, 400W, 800W, 1200W, 1200W
Temperature control range: +5~100°C
Temperature control accuracy: ≤ ± 0.5 ° C
Heating rate: 70 minutes from room temperature to boiling point
Fuse tube: 4A 6A 8A 10A 10A
This product is a tank type, tank made by stainless steel , cold-rolled steel chassis with high-quality well-made. Temperature section with a special electronic temperature control. The pot with digital display actual temperature, the product with beautiful shape and corrosion resistance, stability temperature, energy-saving and practical, easy maintenance, long life and other characteristics.
Using methods and precautions
1. Security of electricity: The product use of three holes safety outlet socket AC 220V power, please properly grounded, please do not ground to gas pipe.
2. Please inject water into the pot above bulkhead before plug the power. Please make sure do not out of the water when using heat pipe, otherwise it may cause burnning heating pipes, leak solder melt or electric shock, etc..
3. Setting power and temperature : Press the key of SET to check the point of temperature. Press button digital control of SET then the character starts to flash, indicating that the instrument into the setting, press the key to set the value increases , decreases. Press key to set the value, press key quickly again to change the instrument back to normal operation temperature when setting is completed. (If the temperature controller is intelligent, then attach the thermostat instructions ).
4. The machine has heating and thermostatic two light indicates working status , thermometers head showing the value of the actual temperature of the pot.
5. Please do not inject too much using water when 100 Celsius boiling distillation, to avoid overflow when boiling.